HOBO Water Level loggers can be hung in wells using Onset's special low-stretch cable, with Teflon coating to avoid snagging and to make them easy to clean.
CABLE-1-50 (SS, 1/16", 50-ft Teflon-Coated SS Cable)
CABLE-1-300 (SS, 1/16", 300-ft Teflon-Coated SS Cable)
CABLE-1-CRIMP (Crimp Sleeve 12-Pack for CABLE-1-XX)
To mount the cable, you will need a secure mounting point at the top:
Mount your wireline to a stainless eye-bolt mounted (drilled and hammered) onto your well cap or well casing.
Pro Hydro well cap:
Secure your logger wireline to a 2-inch-OD (outer diameter) welded stainless ring. It will fit nicely in the 2-inch aluminum well cap and 2-inch male adapter. You must reduce your pipe size to 1.5 inches or 1.25 inches with a reducing bushing right after the well cap. Your ring must have an OD of 2 inches. For example get a 1.5-inch-ID (inner diameter) ring made of 1/4-inch-diameter stainless. One example ring is sold by McMaster-Carr as part number 3769T45 (about $3 each). The 2-inch-OD stainless ring will not pass through the 1.5-inch-ID pipe.
Schedule 40, Reducer Bushing (Flush Style) 2 x 1.25 for 2-inch ring mount
Try dry-fitting the pieces together before you go out into the field. You may need a few other reducers, couplings, cement, tools, etc.
Attach the cable by looping it through the eye-bolt or ring and crimping the loose end to the main cable going to the logger. You will need a crimping tool for this. We recommend using two crimps at each end to be sure that the cable remains firmly attached.