HOBO Pendant Temperature/Light Data Logger 64K - UA-002-64
This miniature data logger can record temperature and relative light levels. Complete with waterproof casing, this product is designed for indoor, outdoor, and underwater deployment. This model can store approximately 52,000 measurments of 10-bit readings.
- Low-cost temperature with alarm indication or light intensity
- Waterproof housing for wet or underwater use
- Data readout in less than 30 seconds via fast Optic USB interface
The UA-002-64 Data Logger is for use in Outdoor and Underwater environments
The UA-002-64 Data Logger supports the following measurements:
Temperature: -20° to 70°C (-4° to 158°F)
Light: 0 to 320,000 lux (0 to 30,000 lumens/ft2)
Temperature: ± 0.53°C from 0° to 50°C (± 0.95°F from 32° to 122°F), see Plot A
Light intensity: Designed for measurement of relative light levels, see Plot D for light wavelength response
Temperature: 0.14°C at 25°C (0.25°F at 77°F), see Plot A
Drift: Less than 0.1°C/year (0.2°F/year)
Airflow of 2 m/s (4.4 mph): 10 minutes, typical to 90%
Water: 5 minutes, typical to 90%
Time accuracy: ± 1 minute per month at 25°C (77°F), see Plot B
In water/ice: -20° to 50°C (-4° to 122°F)
In air: -20° to 70°C (-4° to 158°F)
Water depth rating: 30 m from -20° to 20°C (100 ft from -4° to 68°F), see Plot C
NIST traceable certification: Available for temperature only at additional charge; temperature
range -20° to 70°C (-4° to 158°F)
Battery life: 1 year typical use
UA-002-08: 8K bytes (approximately 3.5K combined temperature and light readings or events)
UA-002-64: 64K bytes (approximately 28K combined temperature and light readings or events)
Materials: Polypropylene case; stainless steel screws; Buna-N o-ring
Weight: 18 g (0.6 oz)
Dimensions: 58 x 33 x 23 mm (2.3 x 1.3 x 0.9 inches)
The CE Marking identifies this product as complying with the relevant directives in the European Union (EU).
NIST testing usually takes between 1-2 weeks.
The following are associated resources for this product. In general users of this product can gain insight into its possible use by utilizing the resources below.