In order to serve you relevant web content, we need to determine your location. If you are located within the United Statesplease click here.
Status: Onset will be closed on Monday, May 29th in observance of Memorial Day. Orders and requests submitted after 2:00 PM ET on Friday may not be processed or confirmed until Tuesday, May 30th.
Wiring Diagram - Connecting a Thermocouple to a ZW node
In order to connect a thermocouple to a ZW, you will need to use a third-party thermocouple transmitter, like the Omega TX91A series, which is available directly from Omega, click here.
In addition you will need the AC-SENS-1 power supply to provide 12VDC to the transmitter. Also needed is the CABLE-4-20MA.
The 4-20mA input to the ZW will need to be scaled in HOBOnode Manager to represent the 4-20mA as temperature in Fahrenheit or Celsius. The appropriate scaling values will be based on the model transmitter you purchase from Omega and the type of thermocouple being used.
Please see the attached drawing for more wiring details.